RGC - Remote Gas Calibrator for Catalytic Bead Sensors
The Remote Gas Calibrator (RGC) allows calibration gas to be applied to a remotely located catalytic bead sensor from an easily accessible location. The RGC is used for blocking ambient air and redirecting methane to the catalytic sensor for calibration or testing sensor accuracy. The RGC tests or calibrates the General Monitors catalytic sensor with 50% LEL methane. The unit is capable of calibrating gas at wind velocities up to 50 mph. The RGC is used manually with a pressure regulator (P/N 80146-1) and is suitable for calibrating with any light hydrocarbon.
RGC-IR - Remote Gas Calibrator for Point IR Detectors
The remote gas calibrator, RGC-IR, is designed to be permanently installed on a combustible IR gas detector. The RGC-IR provides protection from outside elements and allows the user to apply calibration gas (@ 50 psi) to the detector from a remote source.
RGC-HT - Remote Gas Calibrator for High Temperatures
To verify the proper operation of a catalytic combustible gas sensor, it is necessary to periodically apply a gas of known concentration to the sensor. This is usually done manually at the site with gas supplied from a small lecture bottle. As the sensor reacts to the calibration gas, adjustments are made to the monitoring system to bring its calibration into agreement with the known concentration of the calibration gas. At times, sensors are mounted in inaccessible locations, requiring expensive catwalks or scaffolding to reach them. The High Temp Remote Gas Calibrator (RGC-HT) allows the calibration gas to be applied to the catalytic bead sensor from easily accessible locations. The RGC-HT is used for blocking ambient air and redirecting methane or other light hydrocarbon gases to the catalytic bead sensor for calibration or testing sensor accuracy in various environments. The RGC-HT tests or calibrates the General Monitors Catalytic Bead sensor with 50% LEL methane or other light hydrocarbon gas. The unit is capable of calibrating and detecting gases in various wind conditions (up to 45 mph) and temperatures (up to 200°C when used with high temperature sensor).